The first processing level of the LFI DPC is the so called Level 1. The source data of the Level 1 software includes:
- raw housekeeping telemetry packets retrieved from different satellite subsystems: the LFI instrument, the Sorption cooler, the HFI instrument and the Central Data Management Unit (CDMU).
- the LFI raw scientific telemetry
- Additional auxiliary data provided by the MOC and the Flight dynamics:
- The Attitude History File (AHF)
- Time correlation data (time correlation coefficients and time couples)
- The Sorption cooler out of limit data
Only a subset of the raw housekeeping telemetry packets is daily processed and converted into TOIs, i.e. those relevant to the LFI DQR production and the estimation of the LFI instrument systematic effects.
The LFI scientific telemetry
Each LFI radiometer provides two analog outputs, one for each amplifier chain. In a nominal configuration, each output yields a sequence of alternating , signals at the frequency of the phase switch. By changing the phase switches configuration, the output can be a sequence of either or signals.
The conversion from analog to digital form of each radiometer output is performed by a 14 bits Analog-to-Digital Converter (ADC) in the Data Acquisition Electronics unit (DAE). The DAE transforms the signal in the range [-2.5 V, +2.5 V]: first it applies a tunable offset, , then it amplifies the signal with a tunable gain, , in order to make full use of the resolution of the ADC, and finally the signal is integrated. To eliminate phase switch raise transients, the integration takes into account a blanking time, i.e. a blind time in the integrator where data are not considered. The default value of the blanking time is 7.5 . Both the , the and the blanking time are parameters set through the LFI on-board software. The equation applied to transform a given input signal into an output is:
with DAE when applying a selected gain. The values of and are set by sending, through specific telecommands, the DAE Gain Index (DGI) and the DAE Offset Index (DOI) associated to the desired values., is one of 255 possible offset steps from +0 up to +2.5 V and where is a small offset introduced by the
The ADC quantizes the uniformly in the range , so that the quantization step is mV.
(Planck) Low Frequency Instrument
Data Processing Center
(Planck) High Frequency Instrument
Command and Data Management Unit
[ESA's] Mission Operation Center [Darmstadt, Germany]
Attitude History File
Daily Quality Report
analog to digital converter
LFI Data Acquisition Electronics