HFI performance summary
The essential characteristics of the HFI are gathered in the table below. The HFI has been operating flawlessly for 30 months, the maximum autonomy expected for the helium of the dilution cooler. The telescope pointing was achieved with an accuracy better than expected and reconstructed to a few seconds of arc. Fifty of the fifty-two bolometers have collected data in very stable conditions during the whole mission, and practically all of the data was correctly recorded.
The extreme sensitivity provided by the 100 mK bolometers and the HFI architecture has provided a new view of the submillimetre and millimetre sky with a noise not far from the fundamental limits set by the photon noise of the observed source. Expected and unexpected systematic effects were evidenced. The most noticeable of them are the spikes in the signal produced by cosmic rays hitting the bolometers and their immediate surrounding. The overall sensitivity is finally around that announced as the goal in the "bluebook" #planck2005-bluebook, i.e. twice better than the required sensitivity. One can expect to improve these numbers by about root two when the whole mission data is available.
|Frequency||100 GHz||143 GHz||217 GHz||353 GHz||545 GHz||857 GHz|
|Beam size (FWHM, arcmin)||9.66||7.27||5.01||4.86||4.84||4.63|
|Sensitivity per fiducial pixel||3.95 µK/K||2.35 µK/K||4.58 µK/K||15.17 µK/K||0.013 Mjy/sr||0.013 Mjy/sr|
|Goal sensitivity||2.5 µK/K||2.2 µK/K||4.8 µK/K||14.7 µK/K||0.023 Mjy/sr||0.042 Mjy/sr|
(Planck) High Frequency Instrument